33220A LABVIEW DRIVER DOWNLOAD
Most Active Hardware Boards: Modifies the parameters for a square wave, including duty cycle, and outputs a set of three cycles upon receipt of each trigger. So far, I’ve downloaded the drivers for Agilent 33xxx and whenever I initialize the device, it gives me the following error: Agilent A Function Generator Software. But the problem is that “error generated by remote interface command” keeps showing up on the function generator’s screen.
|Date Added:||4 October 2006|
|File Size:||13.11 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Check on this; make sure you have your VIs sitting in a folder, which itself sits in the instr.
Should I try reinstalling Labview. The first one demonstrate how to set the instrument for pre-defined shapes sine, triangle, etc. Consider PXI Arbitrary Function Generators NI PXI Waveform Generators can produce precise waveforms including sine, square, triangle, and ramp as well as arbitrary, user-defined waveforms by using data sequences or by streaming continuously from a host or peer-to-peer instrument within the PXI system.
Now I have a folder saying ivi drivers. How much warranty is with it? Trigger source for this example is the bus. How do you get an MCU design to market quickly? What I am trying to do is very simple.
Why I labvisw getting this substrate picture, when i create a new workspace? They are not to demonstrate the development environment or the IO. Open these examples using the following procedure.
LabVIEW driver for the A | Keysight (formerly Agilent’s Electronic Measurement)
Back to Top 4. They are organized into different zip files, depending upon the programming environment: Select 3220a number of bursts you want the function generator to produce using the burst count control. How much you bought it? Customize the panel of the VI to fit your application needs.
Instrument Control (GPIB, Serial, VISA, IVI)
Agilent A Function Generator Software. This folder needs to be placed directly in the c: Message 4 of I am just using a simple example that NI created agilent 33xxx series generate standard waveform.
Downloads an arbitrary waveform to the generator as binary data. Ultrasonic transducer driver 1. A com port is a VISA resource.
Agilent A Function Generator Software
Don’t understand some of your other statements. Configures a pulse with duty cycle which is then slowly modulated by a triangle waveform.
Right click terminals and choose Change to control to create a control wired to that terminal. Hundreds of built-in functions are included with LabVIEW and the Agilent 33XXX series driver After placing the Configure Burst node, be sure to connect the purple instrument wire and the yellow error wire appropriately.
Agilent Technologies / Keysight Technologies 33220A
But the problem is that “error generated by remote interface command” keeps showing up on the function generator’s screen. Things you can change manually in what is being said in manuals. Am I just dreaming?
Where am I going wrong. Either the com port is listed or it’s not.
Message 10 of The example programs are provided solely for the purpose of illustrating a subset of the SCPI Standard Commands for Programmable Instruments commands used with the A.